We have an optima TLX ultra in our lab and experiencing issue ralating to the CPU error.Is tis a repairable? I tried to cleared it by turning instrument power off and back on.problem still axists.
Identification and quantification in SIM mode
I am using Agilent system for monitring of a novel compound in environmental samples. My compound give exclusively the parent ion and the daughter ions are in trace amounts. Can I use this single ion for identification as well as quantifaction?
PbSmart NIR detector Cary 5000: Ragarding
I want to clarify that,
1. How Pbsmart NIR detector in cary 5000 is superior than the PBS detector in LAMBDA 1050 (PerkinElmer)
2. How PbSmart NIR detector shall better substitute the InGaAs detector for NIR range.
3. Why InGaAs detector...
Run ECD and FID on same sample
Does anyone know how to run a GC system using the ECD and FID on the same sample, i.e. analyze a sample for diacetyl and acetaldehyde on the same vial during a run.
- Last Post: February 4th, 2016
Repairing an Eppendorf 5810
Hi, has anyone done a disassembly and repair of the latch motor on one of these? I'd like to do it myself since Eppendorf quoted us more than we care to spend for a service visit. Any videos or tutorials? I have the repair...