O. I. Analytical
GC starts by itself
When I create a sample list or try to inject a single sample, the equipment starts by itself when it's ready. It doesn't stay "waiting". It goes from "waiting" to "running" automatically, so I must be prepared. It's complicated when I'm trying to do a sample list, because I must be prepared when one sample finishes, to inject the next one, because the equipment starts right away. I'd like to know if someone can help me, because I can't find anything about it in the manuals.