Storing test results
HI! We have purchased a AAS 800 spectrometer about 3 years ago. We did not know until now that it is necessary to store calibration curves and data. So we have tried to store them, but it has failed. How can we store a calibration curve and the data?
PE AA/ICP/ICPMS Tech
Regardless of whether or not you use automated analysis control with an autosampler or manual control, there is a check box on those windows labeled "save data" that must be turned on. There's also an entry box for "dataset" with a browse button next to it that is used to open or create the dataset where data will be stored.