Where can i get some information about connecting AP300 to PC through RS232? I am intending to build database for collecting analysis data, so I need information about UART, message structure etc.
Thank you in advance
Wes - ATAGO
Hello. My name is Wes LeMay Jr. from ATAGO USA Inc. I am in the Technical Customer Service department and will be happy to help.
Connecting the AP-300 to a PC via RS-232C is relatively simple. Once the unit is connected via a RS-232C cable (ATAGO part # RE-75302), that computer just has to have the HyperTerminal set-up to receive the signal. On page 49 of the instruction manual, you will find the appropriate settings.
Page 55 of the manual illustrates how the information is represented on the screen.
If you do not currently have a manual or need additional information, please contact me at firstname.lastname@example.org and I can forward you some more detailed setting instructions. I would also be happy to answer any additional questions that you may have. I look forward to hearing from you soon.
Wes LeMay Jr. (Mr.)
Technical Customer Service
ATAGO USA, Inc.