Serving Bioresearch Worldwide
Centrifuge trips GFCI recepticle
GFCI receptice trips when the unit reaches 7500RPM. Load reading shows a max of 2 amps. There is no out of balance light or error code. Does anyone have any experience with this problem?
The GS15/Allegra21 series is not designed to run on a GFCI outlet, according to Beckman Coulter. I saw this a lot during my career with them.