...a LabX.com Website
Request a Quote for equipment in the GC Systems category and LabWrench can have qualified vendors contact you directly.
Remove From Compare List
The first triple quadrupole with Smart Productivity for high efficiency sample throughput.
Malvern Instruments OMNISEC
The world’s most advanced multi-detector GPC/SEC system
Gow-Mac Series 8100
The Series 8100 gives you sensitive, accurate, reproducible results.
Falcon Analytical CALIDUS micro GAS CHROMATOGRAPH
Falcon Analytical Introduces the CALIDUS microGC Faster, Smaller, Smarter, Easier and Greener than Traditional Gas Chromatographs
GenTech Scientific Master GCxGC TOF Plus
Master GCxGC-TOF Plus - Obtain Improved Productivity and Performance
This Product Has Been Discontinued
Agilent Technologies 6890N
Agilent 6890N GC
Shimadzu GC-8A Series
Simple, Dedicated, Single Detector Gas Chromatographs
Thermo Scientific C2V-200
Micro GC Gas Chromatograph
Cambridge Scientific 300 Series
The 300 series fast GC is an unconventional gas chromatograph using directly heated columns for rapid heating and cool down. It is truly a fast GC
Compare Selected Instruments in GC Systems
Join the GC Systems Community and receive updates on all related articles, forums, and equipment.
Select GC Systems instruments to compare by clicking a
under the product picture.
Please enter all fields
Questions for GC Systems
EFC won't reach injector pressure setpoint
The injector pressure on my 1177 injector won't reach it's setpoint of 15 psi. I've verified that the column dimensions in the setup are correct...
PEAK ARES ARE GETTING LOW
hello gd evening we are using gc-fid but my problem is we are getting low low peak area counts of BTEX,EARLIER WEGOT 50 COUNT NOW WE ARE GETTING 10...
Electrometer failure - Varian CP 3800
Hello! On the GC screen the following message appeared: "Electrometer board EEPROM failure. Please remove and replace the defective middle...
Electrometer board EEPROM failure. Please remove and replace the defective middle electrometer board I got this message on the GC, could you help...
battery voltage out of range
I have received an error code 50 indicating that the battery voltage is out of range. This does not appear to be covered in the manual and I am not...
News for GC Systems
‘Unsurpassed protein SEC’: OMNISEC delivers accurate information with minimal sample
Solving the problem of lignin characterization Malvern presents proven strategies for successful GPC/SEC analysis
Shimadzu Scientific Instruments Announces Product Launches, Presentations and Poster Sessions for Pittcon 2015
About Us |
Getting Started |
Terms of Service |
For questions about using the LabWrench web site please call us.
Toll Free: (888) 781-0328 • International: (705) 528-6888
©2015 labwrench.com, a division of the LabX Media Group. All Rights Reserved.