JM Science/AQV-300 - Dispersing too much iodine
The AQV-300 titrator is dispersing too much iodine in our MEOH solution when a sample is ran. Do you have any ideas for why this may be happening? I have tried to clean the electrode prongs several times and this does not help. This electrode is only 3 weeks old and I am concerned that this is happening so soon. We had a similar problem with our previous electrode that was a year old. Can you please help me troubleshoot this issue. Advise this member.
Bio-Rad/BioLogic HR - Software
Is there any way possible to obtain the software to run a Bio-Rad Biologic HR System? We have a Duoflow system with a bad pump and I was hoping to use the older HR system, but our software doesn't recognise it. Advise this member.
BioTek/Synergy H4 - Having a first read and second read
I am trying to do a Free fatty acid absorbance assay and the first read should be as the blank for the final read. How can I set this in Synergy H4 for the calculation and subtraction of first read from the second one automatically (not manually)??? Advise this member.