The PHASE II SPG-1000 Surface Profile Gauge for sandblasted surfaces only.
Surface roughness, for purposes of this forum, is the result of some machining process and its effect on the material surface in question.
Skid type surface roughness testers profilometers are common instruments used on the shop floor. A diamond stylus is traversed across the specimen and a piezoelectric pickup records all vertical movement. Peaks and valleys are recorded and converted into a known value of a given parameter.
Parameters differ in how they approach looking at peaks and valleys. The most popular parameter is “Raâ€. Ra is commonly defined as the arithmetic average roughness. While the Ra parameter is easy and efficient, there are other parameters that can be more specific and useful depending on the application requirements. It is the parameters that enable us to define surface roughness.
Today, for the purpose of checking Ra values, the use of portable, hand held, surface roughness testers are not only economical, but are digital and easy to use. These surface roughness testers are a given necessity for any shop floor that receives work with Ra requirements.
Active Questions & AnswersAsk a Question
There are no current Discussions
Documents & Manuals
There are no Documents or Manuals available.
There are no Features available.
|Power Requirements||4 AAA Batteries|
+/-5% or 5um (Whichever is greater)
Inch or Metric