Get unsurpassed accuracy with the Pyris 1 TGA
Our TGA embodies the next step in thermogravimetric evolution: simplicity and ruggedness in an easy-to-use, easy-to-maintain, automated system. Never before has there been a TGA more perfectly suited for demanding routine environments and R&D.
Active Questions & AnswersAsk a Question
Recent Questions & Answers
Need Equipment Support?
There are no Features available.
|Temperature Range||50 to 1500 °C|
|Power Requirements||100 to 240 Volt, 50/60 Hz|
A vertical design with a high sensitivity balance and quick response furnace. The balance is located above the furnace and is thermally isolated from it. A precision hang-down wire is suspended from the balance down into the furnace. At the end of the hang-down wire is the sample pan. The sample pan’s position is reproducible.
Static or dynamic, including nitrogen, argon, helium, carbon dioxide, air, oxygen, or other inert or reactive gases.
Analyses done at normal or reduced pressures.
Temperature Range: Subambient to 1000 °C
Scanning Rates: 0.1 °C/minute to 200 °C/minute
Temperature Precision: ±2 °C
High temperature furnace
Temperature Range: 50 °C to 1500 °C
Scanning Rates: 0.1 °C/minute to 50 °C/minute
Temperature Precision: ±5 °C
Tare: Reproducible to ±2 Μg
Sensitivity: 0.1 Μg
Accuracy: Better than 0.02%
Capacity: 1300 mg
High temperature quartz, nichrome, or platinum
Standard Furnace: Platinum or Ceramic with capacity of 60 ΜL
High Temperature Furnace: Platinum or Ceramic with capacity of 250 ΜL
Sample mass range:
Up to 1300 mg
Forced air cooled with an external fan and internal booster purge
Standard Furnace: 1000 °C to 40 °C in less than 15 minutes under normal operation
High Temp Furnace: 1500 °C to 100 °C in less than 30 minutes under normal operation
Operates on Pyris software, fully tested on Windows® operating system.
Hyphenated techniques Compatible with the Spectrum One as well as most other FT-IRs and mass spectrometers (MS).
Fast, thorough and efficient due to reduced furnace volume. Less than 3 minutes to purge the sample area of ambient gases (remove 99% of oxygen) and replace the volume with an inert purge gas. 10 minutes to achieve a 99.99% oxygen-free environment.
Developed under ISO 9000