Discussion Category: Thermogravimetric Analysis
Pneumatic furnace lift out of alignment
I am able to align the furnace correctly around the sample pan on my PE Pyris 1 TGA when the reaction tube is closed, but after I lower the furnace to add sample and then raise it, it comes up completely off center.
How do I realign the lift?
Page 1 of 1