Discussion Category: Titrators
Hello, I have an IC that is spiking the baseline 3 samples in from 1.2 to 800. Then it continues every sample after that, until it is permanently at 800. I have changed : Tubes, Filters, Columns, Guards, Supressor, Eluent, Regenerant and I know it is regenerating because the regerant waste is flowing, and it must flow through the chambers to go to the waste. I am at a loss now.
Page 1 of 1