Discussion Category: Viscometers
please how we can calibrat the viscometer (model# LVDV11+
The "calibration" of your viscometer can be completed following the procedure outlined on the manufacturers website. You will need a bath capable of holding 25.0C +- 0.1C. You will need a spindle that you will perform the calibration with. Finally you will need silicone standard solution in a 500mL low form griffin beaker. Following the manufacturers procedure and calculations and you should be able to determine if your instrument is in tolerance.
Page 1 of 1