Discussion Category: Centrifuge
Unit will no longer power ON
We've had the 5415R for many years. It seems that no one in the lab ever turned it off (it has always been ON). Well, I turned it off and unplugged it the other day in order to clean the rotor.
Now it will not turn back on. I've tried replacing the fuse (slow blow 6.25A, 250 V), but that did not work.
Is there anything else I can check? I did notice that the fuse holder had positions for two fuses, but only one was occupied (I tried putting in two, but could not get everything to fit back in when two were in place).
Thanks for your attention and help!