Discussion Category: Liquid Handling
DANI ALS 1000 adjustment
I have recently purchased a used DANI GC1000 with an ALS1000. I have managed to adjust the autosampler on the sample, wash and waste positions but i can not adjust the vertical position on the FID. As a result the needle does not go down on the detectors cap but stays 7-8 cm above it and injects the sample. Could please someone help?
Page 1 of 1