Discussion Category: ICP Systems
Anybodty will give or share their information about how to deal with this issue. Thanks.
LabWrench is available to help people with specific questions and troubleshooting problems. Could you please tell use the instrument make and model that the power transistor is on, and the LabWrench team will do their best to assist you in solving the issue. This will also help other community members who face similar challenges.
If you have any questions, please do not hesitate to contact me.
I am having the same issue with my Optima 2100 DV. Were you ever able to solve it?
i did check the grounding of my AC-source and it found not standard. Because my company has UPS for this machine and the AC- output of UPS is not standard for the Optima machine so i cut the ground then it solve the problem. The error disappearded and not repeat again till' now.
Page 1 of 1